Electrodynamometer Wattmeter consists of two coils, pressure coil (PC) and current coil (CC). Pressure coil is connected across the circuit. ELECTRODYNAMOMETER. WATTMETER’S. • These instruments are similar in design and construction to electrodynamometer type ammeters and voltmeters. Before we study the internal construction of electrodynamometer wattmeter, it very essential to know the principle of working of electrodynamometer type.
|Published (Last):||17 March 2004|
|PDF File Size:||9.75 Mb|
|ePub File Size:||3.84 Mb|
|Price:||Free* [*Free Regsitration Required]|
Electrodynamometer Type Wattmeter in general, a watt meter is eldctrodynamometer to measure the electric power of a circuit, or sometime it also measures the rate of energy transferred from one circuit to another circuit.
These coils are called the electric current coils of electrodynamometer type wattmeter. Now wattmete reason is very obvious of using two fixed coils instead of one, so that it can be constructed to carry considerable amount of electric current.
The device consists of a pair of fixed coilsknown as current coilsand a movable coil known as the potential coil. Errors may be due to pressure coil capacitance. Construction and Working Principle of Electrodynamometer Type Wattmeter Now let us look at constructional details of electrodynamometer.
This particularly affects low power e. Working Principle of Electrodynamometer Wattmeter.
For each sample, the voltage is multiplied by the current at the same instant; the average over at least one cycle is the real power. Diode detectors are either directly connected to the source, or used with a sampling system that diverts only a portion of the RF power through the detector.
Electrodynamometer Type Wattmeter
Therefore we need to calculate the average of deflecting torque T d. This site uses Akismet to reduce spam. Your email address will epectrodynamometer be wattmetfr. Now the reason is very obvious of using two fixed coils instead of one, so that it can be constructed to carry considerable amount of electric current. The lowest measurable current is given as 0. The moving system comprises a light aluminium vane which move in a sector shaped box.
For high-power measurements, a calorimeter directly measures heat produced by RF power. Air friction damping is used, as eddy current damping will distort the weak operating magnetic field and thus it may leads to error. Leave a Reply Cancel reply Your email address will not be published.
Due to this current proportional to supply voltage flows through the moving coil. Errors in Electrodynamometer Type Wattmeter Following are the errors in the electrodynamometer type wattmeters: Therefore this fixed coil is often called Current Coil or simply CC.
Related pages Measurement of Three Phase Power. It consists of following parts.
Wattmeter – Wikipedia
A limited amount of electric current flows through the moving coil so as to avoid heating. The Electrodynamometer Wattmeter has two types of coils; fixed and the moving coil. Simple meters may be calibrated to meet specified accuracy only for electfodynamometer waveforms. Therefore, a wattmeter is rated not only in watts, but also in volts and amperes.
For this we connect the wattmeter to the circuit as shown in figure below. Wikimedia Commons has media related to Wattmeters. Where, x is the angle.
They can be used for both to measure AC as well as DC quantities as scale is calibrated for both Errors in Electrodynamometer Type Wattmeter Following are the errors in wattmetwr electrodynamometer type watt meters: This may not be specified in the meter’s manual. Electrodynamometer Wattmeter are often used for the measurement of power consumption.
Working Principle of Electrodynamometer Wattmeter | Electrical Concepts
There are two types of coils present in the electrodynamometer. Therefore for higher voltage Potential Transformer shall be used. Therefore they can easily carry heavier current. The sensor element is maintained at a constant temperature by a small direct current.
For AC powercurrent and voltage may not be in phase, owing to the delaying effects of circuit inductance or capacitance. But the instrument will respond to the average deflecting torque over a time period. And the interaction of these two magnetic fields deflects the pointer of the instrument. The resistor controls the current across the moving coil, and it is connected wwattmeter series with it.